to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||17 July 2011|
|PDF File Size:||9.46 Mb|
|ePub File Size:||2.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 29 Operating the instrument Ensure the instrument is switched off and turn it over. Page 90 Quality control recommendation The measurement time varies depending on the test parameter. Insert the code strip smoothly into the test strip guide up to the stop in the direction of the printed arrows.
Make sure the code strip belonging to these test strips is at hand unless the instrument has already been coded with this code strip.
Cobas Accutrend Plus User Manual
Performing a measurement The time needed to evaluate the sample varies depending on the test parameter. If any problems occur during coding, the meter displays an error message see Error Messages chapter of this manual. Setting The Date Instrument settings Setting the date With the next three settings, you first enter the accutred, then the month and finally the day.
Check the expiration date of the test strip. The application area must be completely covered with blood example Aotherwise incorrect values may be obtained. The next time the Set button is pressed, the minutes can be set again with the M button. Display And Symbols The Accutrend Plus instrument Display and symbols Every time the instrument is powered on, you should check the display.
Explanations on flagging this measurement events, ev. Instrument settings ,anual default day flashes. Page 60 Performing a measurement Prepare the lancing device by inserting a new lancet. If a single measured value is stored incorrectly, one dash — will appear in the display. Liquids must not enter the test strip container. Page 65 Performing a measurement A flashing arrow now instructs you to open the measurement chamber flap to apply the blood.
This column contains the illustration of a display This column contains information relating to the screen. Page 66 Performing a measurement The flashing drop symbol above the strip symbol now instructs you to apply the acctrend. Performing a measurement A flashing arrow now instructs you to open the measurement chamber flap to apply the blood. Instrument settings The default month flashes.
Performing A Measurement Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip. Inserting The Code Strip Performing a measurement Inserting the code strip Hold the code strip with thumb and index finger in the white area located at the end of the strip. Page 30 Operating the instrument Cleaning Cleaning A clean optical measuring system is a basic prerequisite for determining precise measured values.
Brief Overview Of The Steps To Be Carried Out Performing a measurement Brief overview of the steps to be carried out Code the instrument once Insert the test strip Apply blood per test strip lot Page 46 Performing a measurement Start measurement by closing the Display of measurement results measurement chamber flap Quality control recommendation Open the measurement chamber flap. Performing a measurement Blood collection Lance the outer side of the fingertip using the lancing device to obtain a large hanging drop of blood.
Introduction Integrated control functions The Accutrend Plus instrument has numerous integrated or available control functions, including the following: Each time you press the Set button, the last stored value of each memory area, if any, is displayed.
Memory Press the Set button to switch between the memory areas of the two test parameters. If too little blood is applied example Bdo not try to spread it or apply a second drop as this may lead to a false measurement.
Page Error messages Display Error meaning and description Error: Acuctrend the instrument Ensure the instrument is switched off and turn it over. Page 8 Contents Instrument settings Brief overview of the instrument settings In addition, you can flag measurements carried out with control liquids as controls. Set date, time and beeper. Note that all handling of blood samples represents a risk of infection.
Comments to this Manuals Your Name. Page 84 Quality control recommendation After switching on the instrument, check the following displays: We recommend that you always leave the beeper enabled. Quality control recommendation Starting the measurement 11 Close the measurement chamber flap. Press the M button to change the currently flashing figure.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Page 64 Performing a measurement When you press the button, the place of the measured value within the memory is displayed here: Page 36 Instrument settings Press the Set button again to save the current setting and go to the next setting. Performing a measurement Switching the code display You can switch between the display of the stored code numbers for information purposes.
Page 69 Performing a measurement Blood application in the instrument: The code strip is required at least when a accutrebd test strip container is opened and before the strips are used.
To print the manual completely, please, download it. Important Notes Performing a measurement Important notes Always … operate the instrument at the acceptable test-specific temperatures see also the test strip inserts: Page 40 Instrument settings The default day flashes.